Web Date: November 26, 2013
A Sweet Solution For Controlling Fluidics In Paper Devices
Expensive laboratory devices and their expert handlers are often hard to come by in developing parts of the world. To help people in those regions run medical diagnostics or check water quality, scientists want to translate the functions of big pieces of equipment into simple and inexpensive paper-based devices.
A team of researchers now reports a way to expand the capabilities of these small analytical tools. They incorporated a dissolvable sugar bridge into a paper microfluidic device that acts as a valve, allowing the researchers to control the quantity of a fluid delivered to and from parts of the paper (Anal. Chem. 2013, DOI: 10.1021/ac4022677).
One type of assay run on paper devices is called a lateral flow test, which works like the basic home pregnancy test. Users add liquid sample to one spot on the device. The liquid then travels through capillary action down a paper strip to another spot containing reagents. A reaction between analytes and the reagents creates a color change, indicating the analytes’ presence.
However, lateral flow devices are pretty limited in what they can do analytically. Researchers such as Elain Fu of Oregon State University want to add new functionalities to the devices, such as the pump controls and valves used in conventional microfluidic devices. By adding in features that allow researchers to control the timing and amounts of liquids delivered, paper devices could look for more than one analyte in a sample or perform assays with more than a single step. For example, with the capability to handle multiple steps, researchers could include a way to amplify detection signals to perform more sensitive tests such as enzyme-linked immunosorbent assays (ELISA).
Fu and her team wanted to develop an automated shut-off switch for liquid flow in paper devices. Such a switch would allow users to deliver a specified amount of a fluid, perhaps containing a reagent for use in an assay, with minimal intervention, Fu says.
She and her team found the solution in the supermarket: a sugar cube. They thought that a sugar bridge, essentially a small rectangular sugar cube, between two parts of a paper device would eventually dissolve as fluid passed through, shutting off flow. By changing the bridge’s size and composition, researchers could control the amount of liquid delivered.
To test the idea, the researchers built a paper device that folds like a greeting card. On one half of the device, there is a reservoir and a paper strip runway separated by a gap. They put the sugar bridge on the device’s other half. Folding the card lays the bridge across the gap between the runway and a feeder strip connected to the reservoir.
They tested two sugars, trehalose and mannose, to make bridges in two widths, 3 mm and 6 mm. Another variable they played with was the material used to make the reservoir feeder strip, which brings liquid from the reservoir to the bridge. They tested nitrocellulose, glass fiber, and polyester. With different combinations of these three variables, the researchers found they could reliably deliver between 10 and 80 μl of dyed water through the bridge before it dissolved. All of the bridges dissolved within about five minutes.
With the proof-of-principle tests completed, Fu next plans to incorporate the sugar bridge into a paper device that can detect malaria.
Carlos D. M. Filipe of McMaster University recently reported a paper device with an “erodible polymeric bridge.” His bridges used different materials but were designed with a similar goal (Lab Chip 2013, DOI: 10.1039/c3lc50762a).
“Both studies provide interesting solutions to the challenge of self-powered, non-electrical off switches,” says Scott T. Phillips of Pennsylvania State University, who develops paper microfluidic devices. “Certainly this is a capability that my group would like to have.”
- Chemical & Engineering News
- ISSN 0009-2347
- Copyright © American Chemical Society